- Sealing time:0.5~2s
- Size of sealing tube:Φ3～Φ6mm
- Supply Voltage:190V～250V, 50-60Hz
Thermo Sealer, SLM-T2008-E
Because the blood, saline liquid drug etc. is gradually taken
special plastic bags as liquid filling, storage, transport and
transfusion, to replace the traditional bottle type infusion. The
bagged liquid drug sealing requires clean non-toxic, non-polluting,
reliable bonding, convenient operation, the blood bag tube sealer
machine is specially designed for this purpose.
a) This sealer is carefully designed after we listen to opinions
from many customers and improve the old-fashioned sealer, finally
succeffully developed after repeated test. Thus the sealer other
than old model has the following features:
b) This machine adopt microprocessory to control heat time,
heating time is from 0.1s~9.9s depends on the tube thickness.
c) The sealing head provide with an elastic auxiliary device, in
order to prevent or reduce hemolysis generated during heat
d) This sealer voltage can be adjusted automatically.It can
normally work steadily when the voltage changes between
190V ～ 250V. It shows more superiority in the area that has larger
2. Technical Parameter:
a) Supply Voltage：190V～250V,50-60Hz
b) Power of Supply：200W
c) Sealing time：0.5~2s
d) Size of sealing tube：Φ3～Φ6mm
e) Lifespan of the electronic bulb: about 1000hours
3. Operating steps:
a) Plug the power,switch on. The power switch is in front of the
upper right corner.
b) Preheated for 3~5 minutes after the sealer is electrified.
Immediately seal tubes after power on will reduce the sealer’s
c) Automatically sealing when tube is placed in the slot of Head
unit. After sealing, heat unit will be automatically tripping off for the
next cycle later.
4. Maintenance and Repair
a) During sealing, pipeline is in a more loose condition, avoid
tight pull or jitter pipeline, or it easy to make tube rupture bleeding.
b) If there is bleeding or exudate phenomenon, the tube should
be resealed. In the sealer off state, remove the stain timely and dry,
to avoid the heat head stuck and damage the photoelectric sensing
c) When test machine should avoid empty (without tube)
heat-sealing operation, prevents the load short circuit, sparking, to
avoid burn the electron tube.
d) Heat sealing head output power will be different due to
environment temperature, and different thickness of tube material.
e) The sealing head unit has been carefully assembled
debugging and in the best working condition, Don’t casually
dismantled it to avoid unnecessary trouble.
|No heat sealing||lnside high-voltage
|Inspect and replace
|Inspect and replace
|The heat sealing head
sensing port blocking
|check and wipe dirt
|Heating head can not
be reset, or reset
|Debris or bloodstains
blocks into the
|Timely clean the
|External interference||Switch off 7seconds
then switch on.